View and Download Hioki BT instruction manual online. BATTERY TESTER . BT Test Equipment pdf manual download. Also for: Bt View and Download Hioki instruction manual online. BATTERY HiTESTER. Test Equipment pdf manual download. comparator tables and send them back to the Pencil The enhanced measurement current in the . HIOKI (Shanghai) Sales & Trading Co., Ltd.: .
|Published (Last):||11 September 2005|
|PDF File Size:||10.96 Mb|
|ePub File Size:||6.38 Mb|
|Price:||Free* [*Free Regsitration Required]|
Click Windows will begin copying the driver files.
Tablet Only for BT In such a case, separate the zero-adjust board from any metal components. Switch off the instrument and remove the test leads. The instrument settings are restored bioki. Page 13 Notation of the This Manual Indicates a prohibited action. Setting the resistance range 2. Primarily for use with lead acid batteries, easy 4-terminal measurement, 2.
Effect Of Eddy Currents Setting Threshold Values for the Comparator Confirm the settings.
Technical Notes No results were found. Model Remote Control Switch When this switch is attached to the test lead, the instrument can hold the values while measuring them. Click Click [Set Clock]. The actual time needed may vary depending on the processing speed of the personal computer used. The resistance warning limit and flashes. Do not use a ballpoint pen or hard-tipped pen, because there is a danger of scratching the surface and corrupting the data.
This will display the reading screen.
HIOKI BT INSTRUCTION MANUAL Pdf Download.
The image shows that the potential gradient is greater near the sources of current. When transporting the instrument, use the original packing materials in which it was hiokl, and pack in a double carton.
OFF indicator flashes when alkaline battery voltage becomes low. Press the MEMO key while the memory function is off. Reset the range correctly. Is the test lead attached correctly? At this point, the memory feature is in effect. It is a strong, wear- resistant lead.
The sensor used in the temperature probe is a thin, precision platinum film. Unusual comparator results Have comparator permissible values been set incorrectly?
operations manual hioki |
This is because these points are where the electric current is in manjal middle of spreading across the metal plate, leading to higher current density. Stop all applications being executed on the computer recommended 3. The specially designed L Pin Type Lead further improves testing efficiency with an L-shaped tip that lets you reach deep into the battery racks to completely probe manuql terminals. Transfer of all units units of data takes approx.
Select the unit in which you want to clear data. Turn off the power of the instrument and remove the cable. When installation is complete, the next dialog will appear. Disconnect the test lead from the battery subject to measurement.
Nothing displayed on screen even after pressing Manusl key This key cannot be used when no data has been saved. Is the test lead disconnected?
Replacing The Fuse 9. This will finalise the configuration process and restart the instrument.
operations manual hioki 3554
Used for setting the number of measurements used in averaging. The actual time needed may vary depending on the processing speed of the computer used. Displayed when setting comparator threshold values. The following figure shows the principle of the AC four-terminal measurement method. Functional Specifications Functional Specifications 8. When sending the instrument for repair, remove the batteries and pack carefully to prevent damage in transit.
Be sure to connect all four terminal: However, mishandling during use could result in injury or death, as well as damage to the instrument.